Fisher Paykel SmartStick infoUSB2 Data Storage and Software
The SmartStick USB from Fisher & Paykel simplifies data retrieval and ensures a hassle free experience. The infoUSB2
merges the functionality of the InfoUSB (Web) and InfoUSB into one system, and is capable of recording up to 365 days of summary compliance and efficacy data, and 7 days of detailed efficacy data. You can easily transfer data from your the USB to your healthcare provider via an internet connection.
- Simple Data Retrieval
- Easy to Install
- Capable of storing large data
Use the data gathered by the infoUSB2 to generate actionable reports that you can send conveniently to your health provider over the web.
What sort of detailed data can the USB stick record?
The USB can records information such as AHI, Pressure, and Leak data.
UPGRADE OF SOFTWARE
1.Make sure the device is not connected to the supply mains.
2.Remove the InfoUSB from the USB Port if present. Refer to the Use and Care manual on how to remove a
InfoUSB from the USB Port.
3.Insert the InfoUSB with the new version of software into the USB Port.
4.Connect the device to the supply mains to power up the device.
5.The16 points of the menu system will appear on the LCD. Rings will sequentially appear around the dots
during the upgrade. When the upgrading process is finished, the Home Screen will be shown.
DO NOT DISCONNECT THE DEVICE FROM THE SUPPLY MAINS OR REMOVE THE InfoUSB FROM THE USB PORT
DURING THE UPGRADING PROGRESS OR AN ERROR WILL OCCUR.
6.When the Home Screen is shown remove InfoUSB with the new version of software from the USB Port.
7.Reinsert the original InfoUSB removed previously.
DO NOT LEAVE THE InfoUSB WITH THE NEW VERSION OF SOFTWARE IN THE USB PORT AS DOING SO WILL
START THE UPGRADING PROCESS EVERY SINGLE TIME WHEN THE DEVICE IS POWERED UP